
Description of item

-Includes a function to check for abnormal cardiac rhythm, which can detect pulse disturbance. When a heart rhythm abnormality (disorder of pulse) is found, we will inform you with a mark.
・The measurement result for one person can be stored up to 200 times.
・Small size and convenient for carrying around on business trips and outings.
General name: Automatic electronic blood pressure monitor (class classification: 167)
Contraindications/Prohibitions:
【warning】
• Use carefully if you have blood circulation in your wrist.
• If you are on dialysis treatment, anticoagulant, antiplatelet, steroid, or have arrhythmia, consult a doctor before use.
• If you are a child or cannot express yourself, use it carefully with the guidance of your doctor.
[Contraindications/prohibitions]
・Do not adjust medication without consulting a doctor.
・Do not measure on the arm during injury or treatment.
・Do not measure on the arm that is receiving intravenous drip or blood transfusion.
Do not use in rooms with volatile anesthetics containing high oxygen density or nitroxide.
・Do not use in public places.
・Do not use near strong electromagnetic fields.


Item specifications:
・Error in cuff pressure display: Pressure ±3mmHg
・Operating environment: Temperature 10~40℃, Relative humidity 30~85%RH
・Error in blood pressure measurement due to clinical performance test: average error with auscultation method within ±5 mmHg, standard deviation within 8 mmHg


Effect/Efficacy:
Non-invasive measurement of systolic blood pressure and diastolic blood pressure to detect abnormal heart rhythm for health management.



Specification Note:
1. When installing the equipment, note the following points.
・Install in a place where it will not get wet.
・Install in a location where there is no risk of adverse effects from atmospheric pressure, temperature, humidity, ventilation, sunlight, air containing dust, salt, or sulfur.
Pay attention to stable conditions such as tilt, vibration, and shock (including during transportation).
・Do not install in a place where chemicals are stored or where gas is generated.
2. Pay attention to the following points before using the device.
・Be careful when using the equipment together, as it may cause incorrect diagnosis or cause danger.
・If you are pregnant, you should monitor your blood pressure frequently because the blood pressure changes greatly.
・If you have arrhythmia, evaluate the measured value in consultation with your doctor.
・Persons undergoing pacemaker treatment should not check the pulse with this product.
3. Pay attention to the following points while using the equipment.
・Measure the wrist at the height of the heart.
・Be careful not to exceed the time and amount required for diagnosis and treatment.
-Continuous monitoring of equipment and patients for abnormalities.
・If an abnormality is found in the device or patient, take appropriate measures such as immediately removing the cuff.
・If you repeat measurement several times, wait at least 15 seconds between measurements.
4. Pay attention to the following points after using the device.
・Do not use liquid or aerosol sprays for cleaning.


Note the following items regarding the storage location.
1. Store in a place where it will not get wet.
2. Store in a place where there is no risk of adverse effects, such as atmospheric pressure, temperature, humidity, ventilation, sunlight, air containing dust, salt, and sulfur.
3. Pay attention to stable conditions such as tilt, vibration, and shock (including during transportation).
4. Do not store chemicals or in a place where gas is generated.
5. Keep out of reach of children and young children.
6. Drop the device, and if the case is cracked, stop the device and never touch the exposed internal parts of the device.
7. If the product breaks down, do not repair it.
8. Do not modify the equipment.
9. Maintenance and inspection
Before reusing equipment that has not been used for a while, make sure that it operates normally and safely before use.


Manufacturing distributor
Tanita Akita Co., Ltd. 
28-1 Shimoda Shimogisoe, Daisen Horomi, Akita Prefecture
Telephone: 0187(66)2122


